Cited 499 times since 1984 (13.7 per year) source: EuropePMC Circulation, Volume 70, Issue 5, 01 November 1984, Pages 812-823 Continuous measurement of left ventricular volume in animals and humans by conductance catheter. Baan J, van der Velde ET, de Bruin HG, Smeenk GJ, Koops J, van Dijk AD, Temmerman D, Senden J, Buis B
An eight-electrode conductance catheter previously developed by us and used to determine stroke volume in dogs was applied in human beings and dogs to measure absolute left ventricular volume quantitatively. For calibration we developed the formula V(t) = (1/alpha)(L2/sigma b)G(t) - Vc, where V(t) is time-varying left ventricular volume, alpha is a dimensionless constant, L is the electrode separation, sigma b is the conductivity of blood obtained by a sampling cuvette, and G(t) is the measured conductance within the left ventricular cavity. Vc is a correction term caused by the parallel conductance of structures surrounding the cavity and is measured in two ways. The first method, applicable in the anesthetized animal, consists of temporary reduction of volume to zero by suction. The second method uses a transient change in sigma b by injection of a small bolus of hypertonic saline (dogs) or 10 ml of cold glucose (humans) into the pulmonary artery. The validity of the formula was previously established for the isolated postmortem canine heart. The predicted linearity, slope constant alpha, and accuracy of Vc for the left ventricle in vivo were investigated by comparing the conductance volume data with results from independent methods
electromagnetic blood flow measurement for stroke volume and indicator dilution technique for ejection fraction (dogs), thermal dilution for cardiac output (12 patients), and single-plane cineventriculography for V(t) (five patients). In all comparisons, linear regression showed high correlation (from r = .82 [n = 46] to r = .988 [n = 20]) while alpha, with one exception, ranged from 0.75 to 1.07 and the error in Vc ranged from 0.5% to 16.5% (mean 7%). After positioning of the catheter, no arrhythmias were observed. It is concluded that the conductance catheter provides a reliable and simple method to measure left ventricular volume, giving an on-line, time-varying signal that is easily calibrated. Together with left ventricular pressure obtained through the catheter lumen, the instrument may be used for instantaneous display of pressure-volume loops to facilitate assessment of left ventricular pump performance.